Experiment Duration: 1 second to 31 days.
Range and operation modes: 350-10,000 ppm
ADC resolution: 14 bit
Resolution: 1 ppm
Max Sample Rate (S/sec): 100
The sensor must be calibrated before every use according to the following simple procedure:
Connect the sensor to the PC through a USB module or to a battery module.
Wait for two minutes to warm the sensor and press the pushbutton switch on the CO² logger sensor for 3 seconds.
A bottle adapted to the sensor is supplied with the sensor.
A connection between your sensors and hardware is required. You can choose from the USB Module, Wifi Module, RF Module, or Display Module. A Battery Module is also required for any connection other then USB Module, which is powered through the cable.