Sensor Calibration: This sensor gives a fast response across the full pH range and can be calibrated with any standard buffer solution. The sensor should be connected to a voltage source (the USB module plugged into a PC , Neulog's™ Display Unit or Battery Unit). Insert the sensor into a pH = 7 buffer and press the sensor's push button for about 3 seconds. The reading is calibrated to 7.
Experiment Duration: 1 second to 31 days.
Range and operation modes: 0 to 14 pH
ADC resolution: 16 bit
Resolution: 0.01 pH
Max Sample Rate (S/sec): 100
A connection between your sensors and hardware is required. You can choose from the USB Module, Wifi Module, RF Module, or Display Module. A Battery Module is also required for any connection other then USB Module, which is powered through the cable.