Neulog pH Sensor NUL-206
The pH Sensor can be used to measure the static pH values of common liquids (water, milk, soft drinks, vinegar, etc.) as well as the changing values in titrations or experiments such as those looking at the effect of antacids.
This sensor can be used to measure the static pH values of common liquids (water, milk, soft drinks, vinegar, etc.) as well as the changing values in titrations or experiments such as those looking at the effect of antacids. The pH sensor is designed for long life in a variety of general purpose situations. Its sealed reference system and gel fill make it easy to use and maintain. With an epoxy body it is a durable electrode for use both in the laboratory and in the field.
Sensor Calibration: This sensor gives a fast response across the full pH range and can be calibrated with any standard buffer solution. The sensor should be connected to a voltage source (the USB module plugged into a PC , Neulog's™ Display Unit or Battery Unit). Insert the sensor into a pH = 7 buffer and press the sensor's push button for about 3 seconds. The reading is calibrated to 7.
Experiment Duration: 1 second to 31 days.
Range and operation modes: 0 to 14 pH
ADC resolution: 16 bit
Resolution: 0.01 pH
Max Sample Rate (S/sec): 100
A connection between your sensors and hardware is required. You can choose from the USB Module, Wifi Module, RF Module, or Display Module. A Battery Module is also required for any connection other then USB Module, which is powered through the cable.
Products being sold are not toys. They are for Educational / Laboratory use only. They are not for use by children 12 and under.